- Testing Machines of Insulators
- Insulators electrical test equipment
- Insulator FRP core oil puncture test machine
Insulator FRP core oil puncture test machine
This testing machine provides a sample test for polymer insulators, it is applied to test the electrical puncture voltage of fiberglass core (FRP/ECR rod) of polymer insulators.
- General Details
- Technical info
- Related Products
What is an insulator FRP core oil puncture test machine?
This testing machine provides a sample test for polymer insulators, it is applied to test the electrical puncture voltage of fiberglass core (FRP/ECR rod) of polymer insulators. Thus the full name of this testing machine shall be “Polymeric insulator FRP rod oil puncture testing machine”.
Why should we perform this test?
According to IEC 61109:2008 standard (equivalent to Chinese standard GB/T 19519-2014), to test the insulation property of the fiberglass core is an essential step to check the quality of a polymer insulator.
How is the FRP puncture test done?
The FRP/ECR rod is be cut into a small sample in a piece, put into transformer oil which has excellent insulation property than air, then the voltage is applied to both ends and increases till the FRP core sample is punctured. The testing machine will record the test procedure and the puncture voltage.
What does this FRP puncture testing machine have?
This machine has the electrical power frequency device, controlling system, measuring device, automatic protection system, electrical poles, all of which enable it to perform the puncture test of polymer insulator rod in oil as per requirement on test methods and acceptance criteria.
Is this FRP puncture test the same as the insulator puncture test?
No, the insulator puncture test (here we are talking about polymer insulator) is another test. The insulator puncture test is to apply voltage on a complete polymer insulator on both ends of fitting like ball & socket and see the voltage that makes it punctured. We have other testing equipment to do that, please check our product showcase for “Electrical testing system for insulators“. While this testing machine offers the FRP sample test, it’s to test the insulation level of its material: FRP/ECR core.
Full name of the test machine: Polymeric insulator FRP rod oil puncture testing machine
Manufacturer: Contune International Co.,Ltd
Voltage for the test: 100kV DC/AC
Capacity for the high voltage: 10KVA
Power supply: 220V
Power consumption: 2KW
Grounding resistance: 0.1oΩ
Computer control: Yes
Language of the controlling system: Chinese / English / Russian
Machine weight: about 500kgs
Water diffusion test device for FRP core
The water diffusion test device is to test the leakage current and withstand property of the FRP/ECR core of composite insulators under AC voltage after boiling in salty water. This test is defined and required in the IEC standard for tests of composite insulators.
Tracking & Erosion Test Machine of Polymer insulators
Tracking & Erosion tester is used for tests of silicon rubber shed and housing of electrical polymeric insulators. It is to test the anti-tracking and erosion ability of its silicon rubber material as per IEC or DIN EN standard.
Insulator Test Equipment for Power Frequency Test
Test equipment for electrical tests of overhead line insulators. This test machine tests high voltage insulators with power voltage and frequency. The power frequency tests are required in IEC, ANSI, or any other technical standards for insulators of OHTL.
33kV Composite Pin Insulators
33KV composite pin insulators are used on 33KV high voltage overhead power distribution lines. These pin insulators are made by FRP core, silicon rubber, and metallic fittings. On the bottom side of the fitting, it is where we call pin, the pin has a screw which is used to fix the insulator onto the cross-arm. On header fitting, it usually has a groove shape to fix conductors.
Fiberglass Rod Of Composite Insulators (Frp/Ecr Core)
Fiber glass rod is used for production of silicon rubber insulator of electrical overhead lines. It is an important part of composite insulator and core material of composite insulator.